Well, some thing happened the week before this, and also now I have an STX, so I want to do some new measurements, measurements that are not RMAA and can be used for comparison against others.
Musiland Monitor 01 USD -> LJM CS4398 -> Fiio E11 -> Xonar Essence STX
Cheap self-solder-terminated-with-Rean-RCA-not-meant-for-SPDIF cable was used in all tests except for jitter, where Belden 1505F terminated with Canare RCAP was used
Sampling rate 44.1kHz unless otherwise indicated
Fiio E11 used to adjust volume such that 0dBFS output gives -0.1dBFS input
|Freq. Response1 10 hz – 19 Khz 24/44||+/- 0.1dB A|
|THD+N 100Hz 0dBFS|
|THD+N 100Hz -0.1dBFS||0.0019% A|
|THD+N 20Hz-1dBFS||See graph A|
|THD+N 10kHz -1dBFS 24/442||0.0014% A|
|THD+N 10kHz -1dBFS 24/96||0.0025% A|
|IMD CCIF 19/20kHz -3dBFS 24/442||0.0007% A|
|IMD CCIF 19/20kHz -3dBFS 24/96||0.0010% A|
|IMD SMPTE -1dBFS||See graph A|
|Dynamic Range -60dBFS||See graph A|
|Crosstalk RMAA3||-95dB A|
|Linearity Error -90dBFS 24/44||0.1dB A?|
|Jitter 11025Hz J-test 24/44||What (See graph)|
1Measured with white noise method in ARTA
2With both input and output sampling rate set to 44.1kHz, ARTA is unable to see frequencies above 22.05kHz
3Fiio E11 not used in this test, volume in Windows set to 93/100 resulting in -1dBFS, actual volume outputted by RMAA unknown
If you think that I'm trying to make a point by intentionally presenting the data in this format, you're right. I've made it so that you can compare against NwAvGuy's measurements of ODAC as much as possible. Some of the numbers cannot be done by ARTA, so I've substituted them with graphs and RMAA when possible. Maximum output line out and Noise dBu are still missing, but if enough people show me that they care, I'll find them out for ya.
Partial results in detail
THD+N 100Hz 0dBFS:
Update: I just discovered that the noise at 0dBFS is due to ARTA. Evidence - even when recording from WAVE I get the same problem, proving that this is totally a software issue. The DAC is not at fault.
So for now, I'll be doing future measurements at -0.2dBFS.
THD+N 20Hz -1dBFS: I decided to not post the numbers according to ARTA, because the frequency resolution is not high enough at low frequencies so I'm suspecting some issues with the THD+N calculation due to huge noise amounts that is thought to exist because the program is unable to see things "sharply" enough at low frequencies. So see the graph instead.
The highest peak, the 3rd harmonic, is below -100dB, which when I last checked is good performance. If you want numbers, I remember it being 0.001x% for THD and 0.005x% for THD+N, so something is definitely wrong with the noise calculation because the graph is mostly noise-free. So it's either the 10k~20k region, or the curve at low frequencies, neither of which is the DAC's fault.
IMD SMPTE -1dBFS: Quite frankly, I don't know how IMD is calculated. But visually, shits are low and people are happy.
Dynamic Range -60dBFS:Noise is low, except for the 50Hz at -11xdB that I still haven't spent time to fix. Maybe if I make a grounded shield for the transformer...
Otherwise, the noise level is low to start with so DNR is nice too.
Jitter: Remember that when you measure something, you are measuring the entire measuring equipment chain. So what I'm measuring here is USB -> Musiland Monitor 01 USD -> SPDIF -> LJM CS4398.
The best part is - LJM CS4398 uses only a CS8416 and nothing else for the digital side (unless you count the DAC chip), and the DAC's external/internal connectors are not 75Ω.